<!--
 * @Author: cgl
 * @Date: 2023-04-18 15:12:26
 * @LastEditors: cgl
 * @LastEditTime: 2023-09-04 14:59:57
 * @FilePath: \public-health-emergencies-fed:\新放射\新radiation-hygiene-fe\src\views\person\itemControl\item08.vue
 * @Description: 
 * 
 * Copyright (c) 2023 by cgl/微盛信息产业有限公司, All Rights Reserved. 
-->
<template>
	<div>
		
		<table border="1" cellspacing="0" cellpadding="0" align="center">
			<tr class="title">
				<td rowspan="2">序号</td>
				<td rowspan="2">检测项目</td>
				<td rowspan="2">检测要求</td>
				<td rowspan="2" width="20%">验收检测判定标准</td>
				<td rowspan="2" width="20%">状态检测判定标准</td>
				<td rowspan="1" colspan="2">稳定性检测</td>
			</tr>

			<tr class="title">
				<td>判定标准</td>
				<td width="10%">周期</td>
			</tr>

			<tr>
				<td >1</td>
				<td >管电压指示的偏离</td>
				<td >口内机，口外机</td>
				<td >±10.0%内</td>
				<td >±10.0%内</td>
				<td >±10.0%内</td>
				<td >六个月</td>
			</tr>
			<tr>
				<td >2</td>
				<td >辐射输出量重复性</td>
				<td >口内机</td>
				<td >≤5.0%</td>
				<td >≤5.0%</td>
				<td >≤5.0%</td>
				<td >三个月</td>
			</tr>
			<tr>
				<td rowspan="2">3</td>
				<td rowspan="2">曝光时间指示的偏离</td>
				<td >口内机</td>
				<td >±5.0%内或±20ms，<br>以较大者控制</td>
				<td >±5.0%内或±20ms，<br>以较大者控制</td>
				<td >±5.0%内或±20ms，<br>以较大者控制</td>
				<td >三个月</td>
			</tr>
			<tr>
				<td >口外机</td>
				<td >$±(5\% + 50ms)$ 内</td>
				<td >$±(5\% + 50ms)$ 内</td>
				<td >$±(5\% + 50ms)$ 内</td>
				<td >三个月</td>
			</tr>
			<tr>
				<td >4</td>
				<td >有用线束半值层</td>
				<td >口内机，口外机</td>
				<td  class="inline" @click="viewInTable('E.2')">
					不低于 <span class="view">查看</span> 规定值
				</td>
				<td  class="inline" @click="viewInTable('E.2')">
					不低于 <span class="view">查看</span> 规定值
				</td>
				<td >——</td>
				<td >——</td>
			</tr>
			<tr>
				<td >5</td>
				<td >高对比度分辨力</td>
				<td >数字成像设备</td>
				<td >≥2.0 lp/mm</td>
				<td >≥2.0 lp/mm</td>
				<td >≥2.0 lp/mm</td>
				<td >六个月</td>
			</tr>
			<tr>
				<td >6</td>
				<td >低对比度分辨力</td>
				<td >数字成像设备</td>
				<td >可分辨0.5mm厚铝板上1mm直径孔</td>
				<td >可分辨0.5mm厚铝板上1mm直径孔</td>
				<td >可分辨0.5mm厚铝板上1mm直径孔</td>
				<td >六个月</td>
			</tr>
			<tr>
				<td colspan="7">
					<div style="text-align: left; margin-left: 20px;">
						<strong>注：</strong>对于含头颅摄影功能的多合一设备，需分别检测全景扫描时和头颅摄影时的高对比度分辨力和低对比度分辨力。
					</div>
				</td>
			</tr>

		</table>

		<el-dialog
			title="牙科X射线设备的半值层"
			:visible.sync="inTableVisible"
			:close-on-click-modal="false"
			class="standarddialog"
			append-to-body
			v-dialogDrag="true"
		>
			<table class="custable" border="1" cellspacing="0" cellpadding="0" align="center">
				<tr>
					<td rowspan="2">序号</td>
					<td rowspan="2">应用类型</td>
					<td colspan="2">X 射线管电压 <br> kV</td>
					<td rowspan="2">最小半值层</td>
				</tr>
				<tr>
					<td>正常使用范围</td>
					<td>所选择值</td>
				</tr>
				<tr>
					<td rowspan="6">1</td>
					<td rowspan="6">采用口内机的牙科应用</td>
					<td rowspan="2">60~70</td>
					<td>60</td>
					<td>1.5</td>
				</tr>
				<tr>
					<td>70</td>
					<td>1.5</td>
				</tr>
				<tr>
					<td rowspan="4">60~90</td>
					<td>60</td>
					<td>1.8</td>
				</tr>
				<tr>
					<td>70</td>
					<td>2.1</td>
				</tr>
				<tr>
					<td>80</td>
					<td>2.3</td>
				</tr>
				<tr>
					<td>90</td>
					<td>2.5</td>
				</tr>
				<tr>
					<td rowspan="10">2</td>
					<td rowspan="10">其他牙科应用</td>
					<td rowspan="2">60~70</td>
					<td>60</td>
					<td>1.3</td>
				</tr>
				<tr>
					<td>70</td>
					<td>1.5</td>
				</tr>
				<tr>
					<td rowspan="8">60~125</td>
					<td>60</td>
					<td>1.8</td>
				</tr>
				<tr>
					<td>70</td>
					<td>2.1</td>
				</tr>
				<tr>
					<td>80</td>
					<td>2.3</td>
				</tr>
				<tr>
					<td>90</td>
					<td>2.5</td>
				</tr>
				<tr>
					<td>100</td>
					<td>2.7</td>
				</tr>
				<tr>
					<td>110</td>
					<td>3.0</td>
				</tr>
				<tr>
					<td>120</td>
					<td>3.2</td>
				</tr>
				<tr>
					<td>125</td>
					<td>3.3</td>
				</tr>
			</table>
			<div slot="footer">
				<el-button @click="inTableVisible = false">关 闭</el-button>
			</div>
		</el-dialog>
	</div>

</template>

<script>
export default {
  name: "item40",
  data() {
    return {
			inTableVisible: false,
    };
  },
  created() {
  },
  methods: {
		viewInTable(type) {
			this.inTableVisible = true;
		}
  }
};
</script>

<style lang="scss" scoped>
@import './content.scss';
</style>